AIRANGER XPL PLUS PDF
This instruction manual provides information specific to the AIRANGER XPL PLUS (Ten Point Level) monitor. When references are made to other Milltronics. AiRanger XPL Plus. * Program range is defined as the empty distance to the face of the transducer (P) plus any range extension (P). **EMC performance. AiRanger XPL Plus. Ten-Point Ultrasonic Level Measurement System. Part 1. General. Scope. A. This section describes the requirements for an ultrasonic.
|Published (Last):||15 November 2017|
|PDF File Size:||5.67 Mb|
|ePub File Size:||1.10 Mb|
|Price:||Free* [*Free Regsitration Required]|
Operation is similar to Space Measurement operation except that measurements are referenced to the transducer face.
To select another transducer Press to set the point number to transducer. A circuit breaker or switch in the building installation, marked as the disconnect switch, shall be in close proximity to the equipment and within easy reach of the operator. If material build-up on the transducer face is observed, to maintain maximum system performance, a schedule of regular cleaning should be adopted.
Page 68 The noise level is a combination of transient acoustic noise and electrical noise induced into the transducer cable or receiving circuitry itself. Introduction After reading this introduction and completing the physical installation, first time system installers may wish to use the AiRanger XPL Plus Quick Start Guide for step-by-step start up instructions. Buy it now – Add to Watch list Added to your Watch list. Obstructed infrared interface, or Clean enclosure “docking bay” defective programmer.
Please enter a number less than or equal to 1. If a Tripper Car is used to fill each section of the bunker, see Example 4 as well. Page P to stabilize the “shot” mode used for the echo evaluation.
Siemens SITRANS LU02 Dual Point Ultrasonic Level Transmitter (Milltronics AiRanger DPL Plus)
Learn More – opens in a new window or tab. For additional information, see the Global Shipping Programme terms and conditions – opens in a new window or tab. Dynamic operating data is retained for one hour and updated immediately on power resumption. Echo Processing Parameters p To P If Empty is substantially higher than the actual vessel bottom, increase Range Extension such that Empty plus Range Extension is greater than the transducer airangger vessel bottom distance.
Relays may be programmed as Standard Alarms or for Custom Relay operation. Press to scroll access the failsafe options. Normal Operation indicates operating conditions are good and the Reading is reliable.
Don’t have an account? SAM required Based on level, space, distance, or volume.
Milltronics AiRanger XPL Plus Instruction Manual
Page 73 To select another transducer Press to set the point number to transducer. If “FAIL” is displayed, repeat the test. May not post to United States – Read item description or contact seller for postage options.
If the peak value jumps wildly, verify the transducer cables are run in grounded metal conduit; No entry required preset for liquid.
Siemens SITRANS LU10 Ten-Point Ultrasonic Level Transmitter (Milltronics AiRanger XPL Plus)
The Offset Correction amount is stored in P Failsafe Operation indicates operating conditions are poor and the Reading may be incorrect. Always prevent the monitored surface from entering the blanking zone.
Table Of Contents Table plud Contents Measurement Parameters P to P Page 74 Enter the number of long shots to be fired and results averaged per transmit pulse. Press with the desired option displayed. Page 18 A circuit breaker or switch in the building installation, marked as the disconnect switch, shall be in close proximity to the equipment and within easy reach of the operator.
Maintenance Transducer inspection, to verify the effectiveness of the self cleaning design, is recommended. An XPL Plus level monitor, utilizing a single Siemens Milltronics ultrasonic transducer, accurately monitors vessel material level without material contact. Point 3 characters, to e.
Milltronics airanger XPL plus Instruction Manual
If Empty is substantially higher than the actual vessel bottom, increase Range Extension such that Empty plus Range Extension is greater airahger the transducer to vessel bottom distance. If Volume Conversion is not required, proceed to Reading Parameters. Enhancement Parameters Scope Displays, P The item you’ve selected wasn’t added to your basket.